Free Shipping to North America, Europe, Australia, Asia by Airmail
30PCS LH Ovulation Test Strips Ovulation Urine Test Strips LH
Tests Strips kit First Response Ovulation Kits Over 99% Accuracy
Item:LH Ovulation Test Strips
Material: Virgin Wood Pulp
30pcs LH Ovulation Test Strips
Immerse the strip into the urine with the arrow pointing towards the urine. Take the strip out after 3 seconds and lay the strip flat on a clean, dry, non-absorbent surface (such as the mouth of the urine container).
IMPORTANT：Do not allow the urine level to exceed the MAX(marker line), otherwise the test may not perform correctly.
The complete reaction time of 3 minutes is required and your result should appear within 3 minutes. Do not read results after 5 minutes.
1. This kit is for external use only. Do not swallow.
2. Discard after use. The test strip is not for reuse.
3. Do not use test kit beyond expiration date.
4. Do not use the kit if the pouch is punctured or not well sealed.
5. Keep out of the reach of children.
How To Use It?
The instructions must be read thoroughly before testing, and the test strips and urine samples should be returned to room temperature (18 ° C - 30 ° C) before use.
1. Remove the test strip from the original packaging foil pouch and use it as soon as possible within 1 hour.
2. Insert the test strip into the urine specimen in the direction of the arrow. Note: The urine level should not exceed the marking line of the test strip.
3. After at least 3-10 seconds, remove it and lay it on a clean and flat surface; or take it out and keep it in the urine cup until the result is read.
4. Wait for the purple band to appear, and the test results should be read within 3-5 minutes. After 10 minutes, the judgment was invalid.
Positive (+): Two red bands appear. One is located in the test area (T) and the other is located in the quality control area (C). And the T line is thicker than the C line, or the C and T lines are both thick.
Negative (-): One or two red bands appear. One is in the test area (T) and the other is in the quality control area (C), but the T line is lighter than the C line, or there is no T line.
Invalid: There is no red band in the quality control area (C), indicating that the incorrect operation process or the reagent strip has deteriorated. In this case, the instructions should be read again and retested with new reagent strips. If the problem persists, stop using the lot number immediately and contact your local supplier.
Note: The ribbon of the reagent strip and the test strip can show the color depth due to the amount of LH in the urine. The result is judged according to the above criteria.
The minimum detectable amount of this product is 10miu/ml.
No cross-reaction with HTSH, HFSH, etc.